Thank you for visiting!
Ceiling Mounted IV Pole
Part# A1 308-100
The IV Ceiling Pole devise swivels, is movable and adjustable. It is anchored to
the ceiling by means of a small steel round structure with bushings and bolts.
The aluminum track swivels 360° clock or counterclockwise. The IV bottle
supporting shaft is telescopic, running from one end to the other, allowing for
the simultaneous utilization of 8 bottles or 4 bottles and infusion pump.
The I.V. Ceiling Pole is permanently affixed to the ceiling and it easily turns
both ways 360° and moves over the bed in either direction on its own 23" track.
The I.V. support pole is completely movable and telescopic (28" to 70"),
allowing it to be locked into position at any height needed by a simple twist of
the wrist. The I.V Ceiling Pole includes the channel track as shown in the
For adjusting the height of the telescopic hook, grasp the upper section firmly
and turn the lower section slightly, to either the left or right, loosening it.
Push the shaft upwards or pull it downwards and choose the desired height. To
lock it up, just repeat the same movement to the left or right.
Comfort and Safety
With more freedom of movement, the patient is able to rest in any chosen
position. The device provides greater safety, removing the risk of accidental
The I.V. Ceiling Pole does not take up any valuable floor space, either in the
intensive care unit or the patients room, facilitating foot traffic and
circulation in the area.
Being anchored to the ceiling, this I.V. pole eliminates direct contact with
children, other nonprofessionals and the danger of passing contagious disease
The installation of the IV Ceiling Pole is very easy. To anchor it to the
ceiling use 2 bolts and bushings. The bushings must be driven into the concrete,
beyond the plaster. For false ceilings, special bushings are utilized.
The device requires no maintenance, except preventive visual maintenance for
verifying the anchoring of the supporting bar. Each six months the anchoring and
firmness of the support must be checked as well. Further, it is necessary to
clean the device as often as necessary. The track must be cleaned with an
alcohol soaked sponge, to make sure no dust or other particles hamper the free
movement of the shaft. This maintenance cleaning is in addition to regular
hospital asepsis care. Replacement of parts such as bolts, hooks etc which are
subject to wear or damage must be done whenever necessary for normal wear,
breaking, corrosion, accident misuse, or any other such defects. The device does
not require any lubrication.
You May Also Like